#jsDisabledContent { display:none; } My Account |  Register |  Help

# Achromatic lens

Article Id: WHEBN0000187884
Reproduction Date:

 Title: Achromatic lens Author: World Heritage Encyclopedia Language: English Subject: Collection: Lenses Publisher: World Heritage Encyclopedia Publication Date:

### Achromatic lens

Chromatic aberration of a single lens causes different wavelengths of light to have differing focal lengths.
An achromatic doublet brings red and blue light to the same focus, and is the earliest example of an achromatic lens.
In an achromatic lens, two wavelengths are brought into the same focus, here red and blue.

An achromatic lens or achromat is a lens that is designed to limit the effects of chromatic and spherical aberration. Achromatic lenses are corrected to bring two wavelengths (typically red and blue) into focus in the same plane.

The most common type of achromat is the achromatic doublet, which is composed of two individual lenses made from glasses with different amounts of dispersion. Typically, one element is a negative (concave) element made out of flint glass such as F2, which has relatively high dispersion, and the other is a positive (convex) element made of crown glass such as BK7, which has lower dispersion. The lens elements are mounted next to each other, often cemented together, and shaped so that the chromatic aberration of one is counterbalanced by that of the other.

In the most common type (shown), the positive power of the crown lens element is not quite equalled by the negative power of the flint lens element. Together they form a weak positive lens that will bring two different wavelengths of light to a common focus. Negative doublets, in which the negative-power element predominates, are also made.

## Contents

• History 1
• Types 2
• Littrow doublet 2.1
• Fraunhofer doublet (Fraunhofer objective) 2.2
• Clark doublet 2.3
• Oil-spaced doublet 2.4
• Steinheil doublet 2.5
• Dialyte 2.6
• Design 3
• References 5

## History

Theoretical considerations of the feasibility of correcting chromatic aberration were debated in the 18th century following George Bass. He realized the two components were for the same client and, after fitting the two parts together, noted the achromatic properties. Hall failed to appreciate the importance of his invention, and it remained known to only a few opticians.

In the late 1750s, Bass mentioned Hall's lenses to John Dollond, who understood their potential and was able to reproduce their design.[2] Dollond applied for and was granted a patent on the technology in 1758, which led to bitter fights with other opticians over the right to make and sell achromatic doublets.

Dollond's son Peter invented the apochromat, an improvement on the achromat, in 1763.[2]

## Types

Several different types of achromat have been devised. They differ in the shape of the included lens elements as well as in the optical properties of their glass (most notably in their optical dispersion or Abbe number).

In the following, 'R' denotes the radius of the spheres that define the optically relevant refracting lens surfaces. By convention, R1 denotes the first lens surface counted from the object. A doublet lens has four surfaces with radii R1 to R4.

### Littrow doublet

Uses an equiconvex crown glass lens with R1=R2, and a second flint glass lens with R3=-R2. The back of the flint glass lens is flat. A Littrow doublet can produce a ghost image between R2 and R3 because the lens surfaces of the two lenses have the same radii. When used in a telescope, it may also produce a ghost image between the flat R4 surface and rear of the tube.

### Fraunhofer doublet (Fraunhofer objective)

The first lens has positive refractive power, the second negative. R1 is set greater than R2, and R2 is set close to, but not equal to, R3. R4 is usually greater than R3. In a Fraunhofer doublet, the dissimilar curvatures of R2 and R3 are mounted close, but not in contact.[6] This design yields more degrees of freedom (one more free radius, length of the air space) to correct for optical aberrations.

### Clark doublet

Uses an equiconvex crown with R1=R2, and a flint with R3≃R2 and R4≫R3. R3 is set slightly shorter than R2 to create a focus mismatch between R2 and R3, thereby reducing ghosting between the crown and flint.

### Oil-spaced doublet

The use of oil between the crown and flint eliminates the effect of ghosting, particularly where R2=R3. It can also increase light transmission slightly and reduce the impact of errors in R2 and R3.

### Steinheil doublet

The Steinheil doublet, devised by Carl August von Steinheil, is a flint-first doublet. In contrast to the Fraunhofer doublet, it has a negative lens first followed by a positive lens. It needs stronger curvature than the Fraunhofer doublet.[7]

### Dialyte

Dialyte lenses have a wide air space between the two elements. They were originally devised in the 19th century to allow much smaller flint glass elements down stream since flint glass was hard to produce and expensive.[8] They are also lenses where R2 and R3 can not be cemented because they have dissimilar curvatures.[9]

## Design

The first-order design of an achromat involves choosing the overall power \phi_{\text{sys}} of the doublet and the two glasses to use. The choice of glass gives the mean refractive index, often written as n_d (for the refractive index at the Fraunhofer "d" spectral line wavelength), and the Abbe number V (for the reciprocal of the glass dispersion). To make the linear dispersion of the system zero, the system must satisfy the equations

\begin{align} \phi_1 + \phi_2 &= \phi_{\text{sys}} \\ \frac{\phi_1}{V_1} + \frac{\phi_2}{V_2} &= 0 \ ,\end{align}

where the lens power is \phi = 1/f for a lens with focal length f. Solving these two equations for \phi_1 and \phi_2 gives

\frac{\phi_1}{\phi_{\text{sys}}} = \frac{V_1}{V_1 - V_2} \qquad \text{and} \qquad \frac{\phi_2}{\phi_{\text{sys}}} = \frac{-V_2}{V_1 - V_2} \ .

Since \phi_2 = -\phi_1 V_2 / V_1, and the Abbe numbers are positive-valued, the power of the second element in the doublet is negative when the first element is positive.

Focus error for four types of lens, over the visible and near infrared spectrum.

## References

1. ^ Daumas, Maurice, Scientific Instruments of the Seventeenth and Eighteenth Centuries and Their Makers, Portman Books, London 1989 ISBN 978-0-7134-0727-3
2. ^ a b c Watson, Fred (2007). Stargazer: the life and times of the telescope. Allen & Unwin. pp. 140–55.
3. ^ Fred Hoyle, Astronomy; A history of man's investigation of the universe, Rathbone Books, 1962, LC 62-14108
4. ^ "Sphaera—Peter Dollond answers Jesse Ramsden". Retrieved July 31, 2009. A review of the events of the invention of the achromatic doublet with emphasis on the roles of Hall, Bass, John Dollond and others.
5. ^ Dokland, Terje; Ng, Mary Mah-Lee (2006). Techniques in microscopy for biomedical applications. p. 23.
6. ^ William L. Wolfe, Optics made clear: the nature of light and how we use it, page 38
7. ^ Kidger, M.J. (2002) Fundamental Optical Design. SPIE Press, Bellingham, WA, pp. 174ff
8. ^ Peter L. Manly, Unusual Telescopes, page 55
9. ^ Fred A. Carson, Basic optics and optical instruments, page AJ-4
This article was sourced from Creative Commons Attribution-ShareAlike License; additional terms may apply. World Heritage Encyclopedia content is assembled from numerous content providers, Open Access Publishing, and in compliance with The Fair Access to Science and Technology Research Act (FASTR), Wikimedia Foundation, Inc., Public Library of Science, The Encyclopedia of Life, Open Book Publishers (OBP), PubMed, U.S. National Library of Medicine, National Center for Biotechnology Information, U.S. National Library of Medicine, National Institutes of Health (NIH), U.S. Department of Health & Human Services, and USA.gov, which sources content from all federal, state, local, tribal, and territorial government publication portals (.gov, .mil, .edu). Funding for USA.gov and content contributors is made possible from the U.S. Congress, E-Government Act of 2002.

Crowd sourced content that is contributed to World Heritage Encyclopedia is peer reviewed and edited by our editorial staff to ensure quality scholarly research articles.